"Family" certification process
In order to reduce costs and simplify the procedures, the Solar Keymark scheme foresees the possibility of certification of a product family, be it a collector or a system.
If the manufacturer produces the “same” collector in different lengths and/or widths
(i.e. the only difference between two collectors is the length and/or the width) the
collector is considered the same subtype (within the same collector “family”).
In this case only one sample of the smallest and one sample of the largest module shall be taken and tested.
Note: the module sizes are compared based on aperture area. Aperture area is defined in EN 12975-2, Annex I. The inspector shall inspect all the different sizes of the type to verify conformity.
The largest module shall be subject to all the tests required in EN 12975-1 clause 5.2, and the smallest shall be subject to a thermal performance test (clause 6 of EN 12975-2). The performance figures used for this type shall be the performance figures corresponding to the measured instantaneous efficiency having the lowest integral in the interval of the reduced temperature from 0 – 0.1 K/(W/m²).
Note: The reduced temperature is defined in EN 12975-2, paragraph 22.214.171.124.3, and the instantaneous efficiency in 126.96.36.199.4.2.
In other words, the efficiency curve used for this subtype shall be the one embracing the smallest area.
If the manufacturer produces the “same” system in different sizes, the different sizes of the system is considered being the same subtype (within the same system “family”). Detailed requirements for systems to be of the same subtype are given in Annex D.
Testing requirements for systems of the same subtype are:
- High temperature and safety tests (according to EN 12976) shall be performed on the sub system having the highest collector area to store volume ratio.
- All other tests (according to EN 12976 and including performance test) shall be performed on the “medium” sub system. Detailed testing requirements are given in Annex D.
Performance indicators for the system configurations which are not performance tested can be determined based on the performance test result on the “medium” system configuration according to the procedures described in Annex D.
Download the latest version of the Annex D:
System families - requirements and extrapolation procedures (version R6 28/3 2011)
Solar Keymark products
Solar Keymark News
PRESS RELEASE - Solar Certification Fund: 8th call launched Fosters quality assurance for tomorrow’s Solar Thermal technology!
Brussels, 18 November 2016 – The Solar Certification Fund (SCF) launched its 8th call for proposals to continue to provide regular and reliable support for (1) developing better and new Solar Thermal certification and standardisation processes, (2) ensuring quality assurance for tomorrow's solar thermal technology.Read more...
The 21st Solar Keymark Network (SKN) meeting took place in the island of Crete, Greece. Among the many points addressed in this meeting, several were related to the outputs of the work linked to the action plan previously approved.Read more...
- Read more...
The 18th Solar Keymark Network (SKN) meeting will be held on 10 and 11 March 2015 in Rome.Read more...
2 October 2014 – The International Code Council (ICC) and the Solar Rating & Certification Corporation (SRCC) have agreed in principle to join forces.Read more...